Plot ? 
Symb ? 
Source * 
OPS? ** 
Average ? 
Thresholds ? 
N_ data 
QC1 parameter 
Data downloads 
Remarks 
method 
value 
unit 
method 
value 
1 
•  QC1DB 
yes (0) 
MEDIAN 
10.7 
ADU 
VAL  0.0,25.0 
1722 
qc_darkmed_ave 
all (full time range)
 median dark level, as computed from the master dark 
2 
•  QC1DB 
yes (0) 
MEDIAN 
0.556 
ADU 
VAL  3.0,4.0 
1722 
qc_darkmed_stdev 
all (full time range)
 standard deviation of median dark level 
3 
•  QC1DB 
yes (0) 
MEDIAN 
3.04 
ADU 
VAL  2.5,3.75 
1722 
qc_ron2 
all (full time range)
 rms noise derived by subtracting 2 consecutive raw darks (normalized to DET.DIT = 1 second) 
4 
•  QC1DB 
yes (0) 
MEDIAN 
2.31 
ADU 
VAL  1.0,4.0 
1731 
qc_gaf_fpn 
all (full time range)
 fixed pattern noise as computed by fitting a Gaussian to the master dark histogram 
5 
•  QC1DB 
no 
MEDIAN 
1.03e+04 
number 
VAL  2000,20000 
1722 
qc_bpm_nbadpix 
all (full time range)
 number of pixels with a high dark current 
6 
•  QC1DB 
no 
MEDIAN 
0.0206 
unitless 
none  
1731 
qc_oes1 
all (full time range)
 
7 
•  QC1DB 
no 
MEDIAN 
0.0183 
unitless 
none  
1731 
qc_oes2 
all (full time range)
 
8 
•  QC1DB 
no 
MEDIAN 
0.0145 
unitless 
none  
1731 
qc_oes3 
all (full time range)
 

*Data sources: QC1DB: QC1 database; LOCAL: local text file  **OPS: OPSLOG data are included

Plot 1 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_darkmed_ave  • 
median:  10.7  ADU 
fixed thresholds:  0.0...25.0  ADU 
N_data plotted:  1722 
[click on plot for closeup] 
Plot 2 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_darkmed_stdev  • 
median:  0.556  ADU 
fixed thresholds:  3.0...4.0  ADU 
N_data plotted:  1722 
[click on plot for closeup] 
Plot 3 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_ron2  • 
median:  3.04  ADU 
fixed thresholds:  2.5...3.75  ADU 
N_data plotted:  1722 
[click on plot for closeup] 
Plot 4 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_gaf_fpn  • 
median:  2.31  ADU 
fixed thresholds:  1.0...4.0  ADU 
N_data plotted:  1731 
[click on plot for closeup] 
Plot 5 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_bpm_nbadpix  • 
median:  1.03e+04  number 
fixed thresholds:  2000...20000  number 
N_data plotted:  1722 
[click on plot for closeup] 
Plot 6 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_oes1  • 
median:  0.0206  unitless 
thresholds:  none  
N_data plotted:  1731 
[click on plot for closeup] 
Plot 7 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_oes2  • 
median:  0.0183  unitless 
thresholds:  none  
N_data plotted:  1731 
[click on plot for closeup] 
Plot 8 scores: not implemented
 data source:  sinfoni_dark (QC1 database) 
dataset:  qc_oes3  • 
median:  0.0145  unitless 
thresholds:  none  
N_data plotted:  1731 
[click on plot for closeup] 
This plot
These are the trending plots for the SINFONI QC parameters derived from DARK frames.
These plots describe:
1. the median DARK level, as computed from each master dark frame.
2. the standard deviation of the median DARK level.
3. the rootmeansquared noise, as derived by subtracting two consecutive raw dark frames, computing a median standard deviation in
selected subwindows, computing a median standard deviation in selected subwindows in the difference image, and then
normalizing to DET.DIT=1 second.
4. the fixed pattern noise, as computed by fitting a Gaussian to the histogram
of the master dark frames. The fpn is simply the standard deviation of Gaussian.
5. the number of pixels with a high dark current.
6/7/8. the oddeven column stripes in the area of slitlet #25. If med(odd) = median of all odd columns in the area of slitlet #25
and med(even) = median of all even columns in the area of slitlet #25 then, this plot is defined as:
oes25 = 2 * (med(odd)  med(even)) / (med(odd) + med(even))
the plots show the effect for the first 3 raw files of the DARK
Find more information about SINFONI DARK frames
here.
Other QC parameters related to detector properties (like dark level,
linearity, gain, etc.) are also trended
here.
General information
Click on any of the plots to see a closeup version.
If applicable, the latest values from Paranal ops logs are plotted as blue dots. More information about those values, including a download option, is available in the OPS column.
The latest date is indicated on top of the plot, data points belonging to that date are specially marked.
If configured,
 statistical averages are indicated by a solid line, and thresholds by broken lines
 outliers are marked by a red asterisk. They are defined as data points outside the
threshold lines
 "aliens" (= data points outside the plot Y limits) are marked by a red arrow (↑ or ↓)
 you can download the data for each parameter set if the 'Data downloads' link shows up
