


QC PLOTS 

LAST 
HISTORY 
conversion factor 


linearity 


QC1 database (advanced users):
browse

plot

Click on CURRENT to see the current (last) trending (Health Check).
Click on HISTORY to see the historical evolution of the trending. 
This page refers to the determination of linearity and gain by the DETMON recipe. Data derived traditionally from imaging screen flats by the fors pipeline can be found here.
To measure the detector parameters a sequence of imaging screen flat field pairs is observed in the R_BESS filter and for the standard readout modes (100Kps/high_gain/2x2 and 200Kps/low_gain/2x2). An appropriate set of bias frames is included in the template.
QC1 parameters
 conversion factor (gain) (QC1 database table fors1_detmon, column gain)
For every pair of flat fields with identical exposure times the difference of the average of the biascorrected frames is compared to the difference in rms between the bias and the flat field difference. This comparison is limited to the illuminated parts of the CCDs. The factor relating the two quantities is the gain in e/ADU. For method=PTC (Photon Tranfer Curve) a linear relation is fit to these quantities for the various exposure times.
 linearity coefficients (QC1 database table fors1_detmon, columns lin_coefi, lin_coefi_err with i=0,1,2)
For each exposure time the mean and the median of the biascorrected flat fields (restricted to the illuminated area of the CCDs) is calculated. Then a second order polynomial is fit to the mean/median vs. the exposure time and the coefficients of that polynomial are stored in the columns mentioned above. In general the contant and the second order terme are 0 within their errors.
 effective nonlinearity (QC1 database table fors1_detmon, column non_linear)
The effective nonlienarity is the difference between the polynomial flux and the one predicted by the linear fir at userdefined flux level (column flux_non_linear), normalized with the polynomial flux.
