With v4.2, calChecker keeps a memory about open NOK/MISS issues beyond the canonical 7 days, up to the configured value for the calibration&issue memory (lower right corner). This is implemented both for 'science' and for 'cal4cal'.
The following keys are used to define a AMBER setup: INS.GRAT1.NAME: PRISM, GMR (medium-resolution), GHR (high-resolution) OCS.OBS.SPECCONF: spectral configuration (Low_JHK, Medium_K_1_2.1 for ex.)
The central wavelength INS.GRAT1.WLEN is also checked to associate the P2VM to the SCIENCE_FRINGE
CALIBRATIONS for SCIENCE:
-- day time calibrations: BPM, FFM should be taken at least once a period
-- night time calibrations:
A P2VM (3P2V or 2P2V) has to be taken at most 6 hours (.25 days) before the science OB,
A calibrator (CALIB_FRINGE) is valid for -1.2/+1.2 hour (.05 day) (see cal4cal)